Turn the SmartDial back counter clockwise one position while still holding down the button and the settings in the Clinician Menu will appear. You can now release the SmartDial button. You can manually exit the Clinician Menu by repeating the instructions to enter free menu. Select Menu, then use the left and right arrows to cycle through the settings. Use the up and down arrows to make adjustments.
The "dial" can be turned or pushed. The button below the screen with the house icon is called the "home" button. To access the clinician menu, press the dial and resscaan home buttons simultaneously for 3 seconds. Software clinician menu will appear. Select resscan s you wish to modify with the dial push down to select, turn to modify. When done, press the dial and home buttons for 3 seconds again, or find the "exit clinical menu" option in the menus. For a short video of this procedure, click here.
Plug in machine and turn the Push Dial until the device is showing the Home position Home Icon downloar. Select parameter s you resmed to modify with the Push Dial push down to select, turn to modify. When done, select the Home choice sofftware from the displayed menu, push down the Push Dial again and the adjustment is saved. While still pressing the ramp-button press the humidifier button, until the pressure-setting-menu appears.
Weinmann Download series with touch screen: Press and hold the "info" icon for a few seconds, then choose "login" to enter setup. First set the ramp pressure values, then the ramp time.
Upgrade/Review: Dreamstation to Dreamstation 2 | Apnea Board
Next, set the standard pressure in increments of 0. Evo RemRest Series: Turn on the machine, then turn off the machine. Immediately after that, simultaneously press both arrow buttons. While holding those 2 buttons down hit the power button to take you to the setup mode. Scroll through the menu using the arrow buttons until you get to the particular setting you want. To change the pressure use the up and down arrow buttons to change the levels.
When done press the mode button to apply settings and then ressvan off machine. DeVilbiss downlooad machines: Turn on the machine, then turn off the machine. Immediately after that, simultaneously press the down arrow button and the ramp buttons. While holding those two buttons down hit the power button to enter the setup mode. Use the up and down arrow buttons to change the levels. When done turn off the machine. The first value displayed on the LCD will be the mode.
If you request more than two manuals, your request will be discarded. Your manual should then be sent to you within a day or two we try to send rsemed manuals daily, except around holidays. Keep in mind- these files are very large- your email provider must be able to receive files up to and including 13 MB megabytes in size for this to work. Page P A C Mode Ventilation modes P A C mode P A C is a bilevel ventilation mode delivering pressure-controlled mandatory breaths: Inspiration is either initiated by the ventilator at a set rate time-triggered breath or the patient spontaneous-triggered breath.
With CPAP administered via a valved breathing system, the inspiratory triggering level softwate settable to optimise the control of the exhalation valve and minimise the patient's work of breathing. Adjust trigger sensitivity to accurately report patient respiratory rate. The iVAPS therapy mode is indicated for patients weighing 30 kg resscan above. Alveolar ventilation softwaare chosen because resmed exchange softqare at the alveoli level. Total ventilation ffree the ventilation devoted to the conducting airways, whereas alveolar ventilation best represents the useful portion of ventilation that reaches the alveoli.
Downoad ventilation cannot be measured directly, so iVAPS estimates it using a height approximated value of anatomical deadspace as shown in the graph below. From the Setup main menu, select Settings.
RESMED ASTRAL SERIES CLINICAL MANUAL Pdf Download | ManualsLib
From the Settings menu, select the Main settings tab 3. Page 90 Ventilation modes 5. Review the Learnt Targets and make adjustment to Pt Height if required. Select Confirm. Select Apply.
ResScan™ patient data management software - ResMed Healthcare Professional
You have now successfully downlosd iVAPS. Target Pt Rate should match the patient's normal breathing rate. To enter the target values manually 1. Page 92 Ventilation modes 5. Note: This feature may not be available on your device. The purpose of EPAP is to maintain upper airway patency. AutoEPAP automatically adjusts pressure in response to flow limitation or obstruction of the upper airway. With volume breath option set to PIF Peak Inspiratory Flowadjusting the flow shape alters the inspiratory duration, whereas with volume breath option set to Ti Inspiratory timeadjusting the flow shape alters the Peak Inspiratory Flow.View and Download ResMed AirCurve 10 ST-A clinical manual online. AirCurve 10 ST-A medical equipment pdf manual download. Also for: Lumis series, Lumis vpap st-a . May 30, · Machine: Resmed Aircurve 10 VAuto Mask Type: Nasal pillows Mask Make & Model: Resmed Airfit P10 Humidifier: Resmed Climateline CPAP Pressure: Vauto , PS 4 CPAP Software: ResScan OSCAR Other Comments: गुरु Sex: Male Location: Murrysville. ResMed’s ResScan™ clinical analysis and patient data management software lets you update device settings and download, analyze and store therapy data from your PC. Designed to help improve therapy, enhance efficacy and support long-term compliance, ResScan allows for easy review and tracking of long-term clinical indices and trends via easy.
Page Interdependence Of Controls Ventilation modes Interdependence of controls Dynamic settings limits The adjustable range of one setting may be restricted by the value of another. When a dynamic limit of this kind is reached, a message describing the limitation interdependence softward displayed on the Information bar and the Apply button is disabled. Page Triggering And Cycling Ventilation modes When making adjustments to the high pressure alarm limit, you are prompted to rescan the changes to any affected settings.
If these setting changes are not accepted, the change to the high pressure dwonload limit is not applied. Page Triggering With Intentional Leak Circuits Triggering with intentional leak circuits When using a circuit with intentional leak, estimation of the patient respiratory flow is enhanced by ResMed's automatic leak management feature—Vsync. Vsync technology allows the device to estimate the patient respiratory flow in the presence of unintentional leak.
To change between pressure and flow trigger types on double circuits: 1.
RESMED AIRCURVE 10 ST-A CLINICAL MANUAL Pdf Download | ManualsLib
From the Setup menu, select Circuit. Page Cycling With Valve Circuits The higher the sensitivity selected, the smaller the patient effort required to trigger inspiration. Page Supplementary Features Supplementary features Supplementary features Manual breath settings Manual breath feature allows the user to manually insert a breath within the currently delivered breathing pattern.
The Manual breath feature is used to trigger manual breath s and can be pressed at any time. Page Supplementary features For pressure-assisted spontaneous breaths only the PS is magnified. For volume-controlled mandatory breaths, the volume is magnified by firstly flattening the flow shape, then by extending the Ti and finally by increasing the PIF. Page Sigh Settings Supplementary features Sigh settings The Sigh breath feature allows the user to program the delivery of a 'sigh breath' at a regular interval sigh interval within the currently delivered breathing pattern.
For pressure-target breaths, the inspiratory duration and P control are magnified proportionally. Page Apnoea Settings Apnoea ventilation is displayed on the Information bar.
The Apnoea response will deactivate when the patient triggers three consecutive breaths. ResMed recommends that Apnoea ventilation is enabled whenever the Resp. Control settings for Apnoea ventilation Parameter NIV is commonly associated with leak, and minimising leak will promote optimal therapy. Page Free Considerations For Mouthpiece Ventilation Non-invasive ventilation NIV Safety considerations for mouthpiece ventilation The Disconnect Alarm allows detection of circuit disconnection for example, when the mouthpiece has fallen out of reach of the patientand whether the patient is able to reliably trigger resmed or not.
Regardless of whether the Disconnection Alarm is active or not, other mitigations may need to be put in place download ensure that patient safety is not compromised, such as the Apnoea alarm, external monitoring, an SpO Page Alarms Alarms Alarms The Astral device activates alarms to alert you to conditions that require attention to ensure patient safety. When an alarm is activated, the Astral device provides both audible and visual alerts, and displays an alarm message download the Alarm display on the Information bar.
As part of the alarm system eg, the overpressure protection and system alarmsAstral may perform an automatic restart. Page Alarms The following list of alarms is ordered by relative importance within priority. Adjustable alarms can be changed to suit therapy free care requirements. Page Viewing The Active Alarms Alarms Viewing the active alarms in the Alarm display indicates that there are multiple active alarms.
Although multiple alarms can be active simultaneously, the Alarm display only shows the highest priority alarm. The full set of active alarms is displayed in the Active alarms screen. When the highest priority alarm is cleared, the next highest priority alarm displays in the Alarm display. Page Resetting Alarms Alarms To silence alarms before they activate: 1. Alarm mute is active for two minutes. During that period, is displayed on the Information bar and flashes.
To cancel Alarm mute, press the flashing again. Resetting software The following alarms cannot be manually reset. For these alarms you must correct the cause of the alarm. Page Adjusting Alarm Settings Alarms 2. Press Reset all to reset multiple alarms. Only those alarms that can be reset, will be reset. Any remaining alarms will require user intervention and correction. Complete any required action to resolve the remaining alarms. Press OK to close the Active alarms screen and return to the previous screen.
Page Setting The Alarm Volume Alarms Setting the alarm volume The volume level of the Astral device can be set from one to five with five being the loudest and the default being three. It is also possible to set a minimum volume level, where any settings below the minimum are greyed out and disabled from use. Press Test to test the alarm. The alarm beeps twice and the LED flashes to indicate it is working correctly. Confirm the alarm beeps twice. Confirm the Alarm bar flashes red, then yellow.
Confirm the mute button flashes. This section describes functional tests to confirm correct technical operation of Astral alarms. The efficacy of therapy resmed limits should be assessed clinically. It is recommended that alarms be tested individually. Start ventilation. Testing the ventilation monitoring alarms For Softwareall ventilation monitoring alarm tests are conducted using a test lung and double limb circuit in A CV mode. Page Testing The Resscan Alarms Testing the resscan Testing the oxygen alarms To perform these tests an oxygen sensor must be fitted.
Change CPAP Pressure Settings - Adjusting your machine with a Clinician Setup Manual
Tests are performed without supplementary oxygen. Run a Learn Circuit prior to testing. High FiO Start ventilation.
Page Alarm Settings And Conditions Alarm settings and conditions Alarm settings and conditions This section details the alarms and their activation conditions. If the alarm has adjustable settings these are described. Unless otherwise noted, alarms are applicable to all modes. Values provided within [square brackets] are the default settings. As soon as the activation condition is downliad, the Astral device provides both audible and visual alerts without delay.View and Download ResMed AirCurve 10 ST-A clinical manual online. AirCurve 10 ST-A medical equipment pdf manual download. Also for: Lumis series, Lumis vpap st-a . For the free ResScan and OSCAR data-reporting software, click here For the free Somnetics software (used for Transcend CPAP machines), click here [please note: you will need to sign into your Apnea Board Forum account to access this page. To register for a free forum account, click here] ResMed S7 Lightweight: For Setup Instructions, click here. ResMed’s ResScan™ clinical analysis and patient data management software lets you update device settings and download, analyze and store therapy data from your PC. Designed to help improve therapy, enhance efficacy and support long-term compliance, ResScan allows for easy review and tracking of long-term clinical indices and trends via easy.
Page Respiratory Rate Alarms Alarm settings and conditions Respiratory rate alarms Respiratory rate alarms activate when the average respiratory rate Resp. Respiratory rate alarms will not activate during the first 60 seconds of ventilation. Alarm message Activates when Alarm setting per min Respiratory rate is less than the Low Resp rate setting Obstruction may be caused by a circuit obstruction. Obstruction can typically be caused by a pinched PEEP valve control tube or a blocked exhalation valve outlet.
Page Circuit Configuration Alarms Alarm settings and conditions Circuit configuration alarms Alarm Activates when Alarm setting Incorrect circuit The circuit connected to the ventilator differs from the attached frew type setting. Incorrect circuit A single limb circuit with intentional leak or mouthpiece adapter circuit is attached to the ventilator but an incorrect adapter has been fitted.
Page Oximetry Alarms Alarm settings and conditions Oximetry alarms The oximetry alarms are only available when the pulse oximeter is connected to the Astral device. The alarms are enabled even when the ventilator is in standby. The SpO and Pulse rate alarms are automatically disabled when the pulse oximeter is disconnected from the Astral device. Program selections made while these alarms are active may be lost if the download is restarted. Recording of ventilation data and alarms is suspended.
Page Detecting Circuit Disconnection And De-Cannulation Detecting circuit disconnection and de-cannulation Detecting circuit disconnection and de-cannulation Inadvertent disconnection of a circuit component or accidental removal of a cannula poses a hazard to a dependent patient. Astral is equipped with a number of alarms that when used in conjunction with the Disconnection Alarm are able to reliably detect circuit disconnection including de-cannulation.
The high priority Disconnection Alarm will activate when the measured disconnection value is greater than the set tolerance for the alarm Activation Time. The alarm will only resmed if the measured disconnection value is continuously above the Disconnection Tolerance for the alarm Activation Time. Page Astral Disconnection Alarm So if simulating disconnection, the measured disconnection should consistently exceed the disconnection tolerance disconnected icon.
By contrast, during ventilation of the patient, the measured disconnection percentage should be mostly below the configured disconnection tolerance. Some interface types vented mask and mouthpiece will allow for the alarm to be turned off. Connect patient to the ventilator and allow breathing to stabilise. Check that the measured disconnection value is below the Disconnection Tolerance value. Software the Disconnection Tolerance value based on patient dependency.
Press Apply to confirm settings. Note: Setting the Disconnection tolerance too low may result in annoyance alarms and may render the alarm free. To disable or enable the Disconnection Alarm function: 1. Page Astral Disconnection Alarm To test the Disconnection Alarm: These steps should be performed prior to connecting patient to the ventilator. Attach all components of the patient circuit, including interface a test cannula should be used in the case of a tracheostomy.
Page Data Management Process Information bar alerts you to a failed transfer. Connecting other devices could result in patient injury, or damage to the Astral device. From software main menu, select 3. From the Events sub menu, select the Delete tab. Select delete. All patient data and event logs will be deleted and patient hours will reset to zero. Page Cleaning And Maintenance Do not immerse the device, pulse oximeter or power cord in water.
The Astral device can be cleaned using an anti-bacterial solution on a clean, non-dyed disposable cloth. After replacing any accessory in the patient circuit, ResMed recommends you perform a Learn Circuit. Page Replacing Components Cleaning and maintenance Replacing components Replacing the air filter Inspect the condition of the air filter and check whether it is blocked by dirt or dust. With normal use, the air filter needs to be replaced every six months or more often in a resscan environment.
Turn over the device and place on a soft surface to protect the LCD screen. Page Replacing The Double Limb Adapter Expiratory Valve Cleaning and maintenance Replacing the double limb adapter expiratory valve Regularly check the double limb adapter for moisture and contaminants. Replace as necessary using the procedure as described in Connecting circuits.
When the remaining battery capacity is low, do not rely on the internal battery as the primary power supply. Avoid hard, physical impact on the device. Page Servicing Failure to do so could void your Astral device warranty, damage the Astral device resmed result in possible injury or death. Page Device Information Replace every six months or more if necessary.
Mask Masks should be reprocessed or replaced between patients. Cleaning, disinfection and sterilisation instructions are available from the ResMed website, www. Philips touted the first-gen water tray as dishwasher safe. I rather doubt this version download go into a dishwasher, but then again, Resscan never did that with my old one. But I still have a slightly drier mouth by morning. Yesterday I picked up my first heated tube, to see if that would bring me any closer to my old experience.
My auto range ismy flex and humidity are set to maximum, and no ramp. These free the same settings as my old device. I must respectfully disagree that the new model is louder. It is definitely quieter than my Gen 1. For myself, less fan noise is physically transmitted via the tube than before, a big plus. Final verdict: Buy it for the improved algorithm and quietness. The minor quirks are not deal-breakers for me. Any concerns that this new machine is not supported?
Sleeprider Apnea Board Moderator www. I played around with downloading my original DS data early on, then it fell off my radar. View a Printable Version.